Operating System / Target
Max Baud Rate
VISA is a high-level API used to communicate with instrumentation buses. It is platform independent, bus independent, and environment independent. In other words, the same API is used regardless of whether a program is created to communicate with a USB device with LabVIEW on a machine running Windows 7 or with a GPIB device with C on a machine running Mac OS X.
USB is a message-based communication bus. This means a PC and a USB device communicate by sending commands and data over the bus as text or binary data. Each USB device has its own command set. You can use NI-VISA Read and Write functions to send these commands to an instrument and read the response from an instrument. Check with your instrument manufacturer for a list of valid commands for your instrument.
Starting with version 3.0, NI-VISA supports USB communication. Two classes of VISA resources are supported: USB INSTR and USB RAW.
USB devices that conform to the USB Test and Measurement Class (USBTMC) protocol use the USB INSTR resource class. USBTMC devices conform to a protocol that the VISA USB INSTR instrument control resource class can understand. No configuration is necessary to communicate with a USBTMC device. To communicate with a USBTMC instrument, refer to the Using NI-VISA to Communicate with Your USB Device section. For more information about the USBTMC specification, refer to the USB Implementers Forum Web page linked below.
USB RAW instruments are any USB instrument other than those instruments that specifically conform to the USBTMC specification. If you are using a USB RAW device, follow the instructions in the Configuring NI-VISA to Control Your USB Device section to configure NI-VISA to control your device. Contact your instrument manufacturer for details about the communication protocol and the command set your instrument uses.
For specific information about the NI-VISA API, refer to the NI-VISA Help. This document is included with NI-VISA and is available through the link at the end of this tutorial.
This section walks through the steps for configuring a USB RAW device to be controlled by NI-VISA 5.0 on a Windows-based computer. If you are using a USBTMC-compatible device, connect your device and skip to the third step, Test the device with NI-VISA Interactive Control.
At this point, NI-VISA already should be installed on your computer, and your USB device should not be connected. Furthermore, you should not have a driver for your USB device installed. There are three steps to configure your USB device to use NI-VISA:
For the purposes of this tutorial, a Creative Webcam is used as an example USB device and is installed on a Windows 7 system. Because this tutorial is intended to explain the configuration of a generic USB device, details specific to the Creative Webcam are not discussed.
To use NI-VISA, you must first tell Windows to use NI-VISA as default driver for the device. In the Windows environment, you can do this with an INF file. NI-VISA 3.0 and higher includes the NI-VISA Driver Wizard to create an INF file for your USB device.
1. To open the NI-VISA Driver Wizard, select Start»All Programs»National Instruments»VISA»Driver Wizard. Figure 1 shows the open screen.
Figure 1. NI-VISA Driver Wizard Hardware Bus Selection Window
You can use this wizard to create an INF file for a PXI/PCI, or USB device. Because you are creating the driver for a USB device, click USB and Next. The NI-VISA Driver Wizard USB Device Selection window opens as shown in Figure 2.
Figure 2. NI-VISA Driver Wizard USB Device Selection Window
2. For this step, you must know the USB vendor ID and product ID for your USB instrument. These numbers identify your USB device when you install it and address your device when you want to communicate with it. According to the USB specification, both numbers are 16-bit hexadecimal numbers and should be provided by the device manufacturer.
If you have access to your USB device, go ahead and connect it to your computer at this time, but cancel out of the Found New Hardware Wizard if it starts. Complete step 2a.
If you do not currently have access to your USB device, but know your USB vendor ID and product ID complete step 2b.
If you do not currently have access to your USB device and do not know your USB vendor ID and product ID, obtain that information from the manufacturer, then complete step 2b.
2a. Open the Device Manager from the Control Panel and find your device on the list, usually under "Other Devices." It may show a yellow exclamation mark indicating it is an unknown device. Double-click the device to open the properties. Select the Details tab and ensure that "Hardware Ids" shows in the attribute drop-down box. A string of characters will be displayed similar to Figure 3. The four characters to the right of "VID_" and "PID_" are your vendor ID and product ID, respectively. Write down the characters for your device, close the Device Manager, and unplug the device from the computer.
Figure 3. Finding the Hardware IDs from the Device Manager
For this Creative Webcam, the vendor ID and product ID are 0x041E and 0x4036, respectively. Note that the vendor ID and product ID will be different for your device depending on the manufacturer and model.
In the NI-VISA Driver Wizard USB Device Selection window use your vendor ID and product ID to match to one of the items in the Device List. Once you've clicked on this device, you should see the other fields in the window populate with information about your device. Verify that this information looks correct. If you do not see a match for your device under Device List, first try clicking Refresh, and if that still does not work then skip to step 2b.
Click Next. The Output Files Generation window is displayed as shown in Figure 6.
2b. In the NI-VISA Driver Wizard USB Device Selection window, choose Other from the Device List.
Click Next. The following prompt shown in Figure 4 will recommend you to connect your USB device.
Figure 4. Prompt recommending you to connect your USB device
Click Yes. The USB - Device Information windows is displayed as shown in Figure 5.
Figure 5. NI-VISA Driver Wizard USB - Device Information Window
Enter the information you've collected about your device into the four fields shown in Figure 5.
Click Next. The Output Files Generation window is displayed as shown in Figure 6.
Figure 6. NI-VISA Driver Wizard Output Files Generation Window
The USB Instrument Prefix is simply a descriptor that you will use to identify the files used for this device, and you can choose any prefix you would like. Enter a USB instrument prefix, select the desired directory in which to place these files, and click Next. The Installation Options window is displayed as shown in Figure 7.
Figure 7. NI-VISA Driver Wizard Installation Options
The default selection is to install the generated files on your computer and is usually the best option. Once you select an option, click Finish to exit the wizard. The INF file is created in the directory you specified in the output file directory field in the previous window.
The installation of INF files is different for each version of Windows. Because of the differences between Windows 2000/XP and Windows Vista/7 the NI-VISA Driver Wizard will create two an INF files, one for each OS group. Installation instructions are included in a header at the top of the INF file. Because INF files are ASCII text files, they can be read in any text editor such as Notepad. For detailed information about installing your INF file, open your INF file in a text editor and follow the instructions at the top of the file. This tutorial assumes you are using Windows 7.
Note: In some cases, Windows may already have a default driver associated with your USB device. If this is the case, Windows will install that driver first. Once you've plugged in your USB device and Windows has installed the default driver, open the Device Manager from the Control Panel. Once Device Manager is open, expand the tree category appropriate to your device, ie: Human Interface Devices. Then find out which instance of "USB Human Interface Device" (shown in Figure 5) corresponds to your USB device by right-clicking and selecting Properties and checking the Details tab for the matching VID and PID as detailed in the Create the INF File Using the Driver Development Wizard section above.
Once you've found the "USB Human Interface Device" that has the matching VID and PID of your USB Device, right-click on it and choose Update Driver Software.
On the first screen, select Browse my computer for driver software. On the second screen, select Let me pick from a list of drivers on my computer. On the third screen, shown in Figure 8, click Have Disk. Direct the prompt browse to C:\WINDOWS\INF, and choose the INF file you copied there and click OK. Make sure your device is selected in the window shown in Figure 8, then click Next. Windows may issue a warning about not being to verify the driver, choose to install the driver anyways. When the driver has finished installing click Finished.
Figure 8. Select the driver for Your USB Device
1. Open Measurement & Automation Explorer (MAX). Select Tools»Refresh to refresh the view to ensure your device appears. Your USB device should be listed under Devices and Interfaces as shown in Figure 9. Your USB device is now installed and configured to use NI-VISA.
Once you've selected your USB device, the device information can be displayed by clicking the USB Settings tab at the bottom of the right pane. Using this window, you can access information such as the manufacturer ID, model code, and serial number for your device.
Figure 9. USB Device Shown in MAX
2. To communicate with your device using VISA, use the VISA instrument descriptor for your device. The instrument descriptor format for a USB INSTR device is USB[board]:: manufacturer ID:: model code:: serial number[:: USB interface number]::INSTR. The instrument descriptor format for a USB RAW device is USB[board]:: manufacturer ID:: model code:: serial number[:: USB interface number]::RAW.
According to the USBTMC specification, all USBTMC devices must have a serial number. Some USB RAW devices may not have serial numbers. If your device does not have a serial number, NI-VISA automatically assigns a VISA specific serial number for that device, as shown above in Figure 9. The format for the serial number is NI-VISA-#, where # is an automatically generated number.
Some USB devices have multiple interfaces. This is similar to the way a PCI device can have multiple functions. If your device only supports one interface, you do not need to include the USB interface number.
The Creative Webcam in this example uses the RAW class, and the manufacturer code and model code are 0x041E and 0x4036, respectively. For the Creative Webcam, the instrument descriptor is USB0::0x041E::0x12C0::NI-VISA-20002::RAW.
To test communication with this device, open MAX. Select Tools»NI-VISA»VISA Interactive Control. A window similar to that shown in Figure 10 should open.
Figure 10. VISA Interactive Control
3. The VISA Interactive Control (VISAIC) is a utility program used to communicate easily with any VISA resource. After your USB device is configured to use VISA, it should be listed in the USB branch. Double-click on your device to open a VISA session to the device. The window shown in Figure 11 should open.
Figure 11. VISA Interactive Control Open VISA Session
When you open a VISA Session with VISAIC, the Configuration section and the USB Settings tab are automatically selected. To read the properties of your device, select the View Attributes tab. This lists the different attributes of your device, including information like Resource Name.
For more information about using VISAIC, refer to Developer Zone: VISA Interactive Control (VISAIC). For information about the NI-VISA API, review the NI-VISA Help. Both are available through the links at the end of this tutorial. For a list of valid commands for your USB instrument, contact your instrument’s manufacturer.
This section explains how to communicate with your USB device using NI-VISA 3.0 and above. Recall that there are two classes of USB devices. The method of communication depends on the class of your device.
Devices that conform to the USB Test and Measurement Class (USBTMC) use the NI-VISA USB INSTR class. These devices use 488.2 style communication. For these devices, you can simply use the VISA Open, VISA Close, VISA Read, and VISA Write functions in the same way you would if you were communicating with GPIB instruments.
Figure 12 illustrates a LabVIEW VI that communicates with a USBTMC device. In this example, a VISA session is opened to a USB device. A command is written to the device, and the response is read back. In this example, the specific command being sent is the ID query for the device. Check with your device manufacturer for your device command set. After all communication is complete, the VISA session is closed.
Figure 12. USBTMC LabVIEW Example Block Diagram
Communicating with the USB RAW class is more complicated because each device may use its own communication protocol. Contact your device vendor for details about the communication protocol for your device.
USB communicates using four types of pipes or endpoints: control, bulk, interrupt, and isochronous. Each type of pipe transfers a different type of information. Furthermore, any number of endpoints can be of any endpoint type. Think of an endpoint as a communication socket. For specific details about USB architecture, review the USB specification linked at the end of this article.
NI-VISA supports three types of USB pipes: control, bulk, and interrupt. When NI-VISA detects your USB instrument, it automatically scans your instrument for the lowest available endpoint for each type.
When VISA detects the lowest available endpoint, it assigns that value to the appropriate VISA attribute. The bulk in endpoint and bulk out endpoint are stored in the VI_ATTR_USB_BULK_IN_PIPE attribute and the VI_ATTR_USB_BULK_OUT_PIPE attribute, respectively. The interrupt in endpoint is stored in the VI_ATTR_USB_INTR_IN_PIPE attribute. A value of -1 indicates that a USB device does not support that type of pipe. For the control pipe, only endpoint zero is supported. If you are using the C API, use the viSetAttribute function to change endpoints. In LabVIEW, use a Write VISA Property node.
NI-VISA includes four functions to transfer data through USB pipes. Before you can communicate with your device using these functions, you need to set up the communication protocol using the VISA USB attributes. The following list describes the available functions.
If you are using LabVIEW, VISA includes an additional function to use the interrupt pipe: VISA Get USB Interrupt Data. In the C API, you can do this by accessing the VI_ATTR_USB_RECV_INTR_SIZE and VI_ATTR_USB_RECV_INTR_DATA attributes of the VI_EVENT_USB_INTR event object. See the NI-VISA Help for more information about VISA Events.
NI-VISA relies on a Linux kernel feature for its USB support. This feature is called usbfs, and on older Linux kernels was referred to as usbdevfs. For NI-VISA to support USB devices, this feature must be present and mounted (like a virtual file system). This is supported in most major Linux distributions such as Red Hat, SuSE, and Mandrake. You may use the mount command to display what file systems are currently mounted to see if your system currently supports this feature.
Also, the VISA user must have write access to the file that represents the USB device, which is typically somewhere in a subdirectory within /proc/bus/usb. If this is not the case, the USB device is not accessible by VISA (it will not be found using viFindRsrc, and viOpen will fail). The default configuration on most systems is that the root user has write access; however, no other user has this access.
There are several options for providing a nonroot user access to a USB device.
As long as no other driver on the system claims the USB device, you can use NI-VISA to access it. No special setup is required.
Return to the Instrument Control Fundamentals Homepage
Developer Zone: VISA Interactive Control (VISAIC)
Developer Zone: Using the VISA Driver Development Wizard and NI-VISA to Develop a PXI/PCI Driver in Windows
Products and Services: Instrument Control Software
Products and Services: Ethernet/LXI
Linux® is the registered trademark of Linus Torvalds in the U.S. and other countries.